Glove Box Every Day Servicing Tutorial
one) Fill in the use record in time right after operating the glove box, which include "user, water oxygen price, what to do"
2) When replacing gas cylinders in the glove box, innovative technology glovebox the pressure is established to the default benefit ± 3mbar
3) The day-to-day working force is set according to the working ease and comfort. When not in use for a prolonged time, the tension is set to 1-6mbar
four) Frequently test the wear of the gloves, look at the oil stage of the vacuum pump (amongst the upstream and downstream markings), and include the pump oil appropriately
five) Clean up the drinking water probe as soon as every 2 to 3 months to make certain exact detection. (Also need to refer to the genuine scenario of the surroundings in diverse user packing containers
six) If you frequently use organic and natural solvents in the glove box, it is recommended that you frequently replace the organic solvent adsorbent product, not later than one 12 months at the most up-to-date
seven) Exchange the filter aspect routinely, no afterwards than two yrs
8) In accordance to the real use of your glove box, determine to exchange the vacuum pump oil cycle, these as black and yellow pump oil, h2o or extra natural and organic solvents in the pump oil. The oil replacement cycle must not exceed 2 many years at the most recent
9) If a significant volume of natural and organic solvent is inevitably exposed during the experiment, the cycle desires to be stopped for the experiment. Following the experiment, first clear and then start off the cycle
10) When the setting in the box is toxic or harmful gasoline, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and the exhaust port of the cleaning valve must be exhausted to the outside with an exhaust pipe (when exhaust is not authorized to be outdoor, the exhaust gasoline treatment need to be finished first).